NOTE: PSS500 NON-HEALTHCARE SOFTWARE ONLY
Select the automatic vacuum leak test and the parameters
- Select CYCLE 8
- Push MENU
- Set VAC ENDPOINT to 24 inches
- Hit ENTER
- HIT ADV
- Close the door and start the cycle
- Pass/Fail status will show at the end of the printout
Note: Allowable leak rate is set in CAL MODE
The automatic vacuum leak test is used to test the integrity of the sterilizer chamber and piping by testing the leak rate of the sterilizer. The test is performed to assure that the sterilizer is capable of holding a vacuum with less than 1 mmHg loss of vacuum per minute over a twenty (20) minute period of time.
Note: In a closed vessel at 30 mmHg, the pressure changes by approximately 0.75 mmHg for each 10ºC change in temperature over the range of 20ºC to 140ºC (65ºF to 284ºF); at 50 mmHg, the change is approximately 1.5 mmHg. The test could be compromised if the temperature changes more than 10ºC during the period in which the chamber pressure is monitored.
Have a question or having difficulties? Talk to Dave Schall, our PRIMUS sterilizer expert who is here to assist you anytime at 877.679.7800 extension 1212 or email at email@example.com.
Visit us at spire-is.com