Take a look at these points when troubleshooting the calibration on a PRIMUS sterilizer.
The chamber was incorrectly calibrated:
Low setpoint 60.9 – should have been 90
High setpoint 60.9 – should have been 135
The load probe had not been calibrated
Still having trouble? Contact our PRIMUS sterilizer expert, Dave Schall at 877 679 7800.